Find great deals for Kodak DryView Laser Imager Xray Film Processor Imaging System. Shop with confidence on eBay!. Find great deals for A Kodak DryView Laser Imager. Shop with confidence on eBay!. Kodak DryView Laser Imager with Kodak PACS Link Systems. This product implements all transactions required in the IHE Technical.
|Published (Last):||17 August 2016|
|PDF File Size:||2.59 Mb|
|ePub File Size:||18.5 Mb|
|Price:||Free* [*Free Regsitration Required]|
Moving Stored Images — Keypad Do not use this product near water, for example, near a bath tub, wash bowl, kitchen sink or laundry tub, in a wet basement or near a swimming pool. Loose clothing, jewelry or long hair may cause minor personal injury or damage to the equipment. Only qualified service personnel may perform installation and service maintenance.
D Keypad – – Imager Menu The numbered descriptions correspond to the numbered callouts in Figure dryvieew D Moving Dryvies Images – – Keypad The move function lets you move a previously stored image to any available non-stored position dfyview. The film is exposed by a laser beam and then fed back into the film transport. Check for and remove any torn pieces of cloth. Artikel wurde bereits benutzt. Use a flashlight and fingers to remove large particles. Used to move from one parameter to the next.
Preisvorschlag senden dtyview Laden This dust can create artifacts on printed images. The intent is to let the operator schedule a convenient time for preventive maintenance and cleaning by a certified technician. The Kodak DryView laser imager is the right choice for health imaging facilities that need diagnostic-quality images but do not require high-volume output.
NISL KODAK DRYVIEW For Sale
By using the ‘Select a language’ button, you can choose the language of the manual you want to view. K Insert a different film cartridge and press the calibration button 87000 the local panel. Get high performance and superb image quality for diagnosis, image sharing, and film archiving.
This button is not displayed if the keypad is connected to a Standard or Plus system. Error Error messages indicate an error condition within the imager. From the local panel, press Supply to close the film cartridge and open the supply door.
Laser Imager (Kodak MRI DryView 8700)
Perhaps the users of ManualsCat. Enhanced Options — Configurable and Upgradeable Design an right for you – Choose one, two, or three film drawers, with an optional six-bin film sorter with a choice of a mammography upgrade – Print high quality mammography images up to 3. Once all the desired images are erased, press Erase Image to exit the Erase Image mode.
Output Button — Selects the desired output destination Imager 1 or Imager 2. View a manual of the Kodak DryView below. Receive Tray — Collects developed film.
Kodak Dryview 8700
Therefore, it must not be used in locations where it can come in contact with liquids, including bodily fluids. Dashed lines indicate the film path.
Press Print to initiate a contrast test film print. Open the platen access door turn the handle counterclockwise and push in the door. Keypad Error Message Table There may be a risk of electric shock from lightning.
Used to decrease dryivew contrast or density setting, or to initiate printing of a contrast or density test sheet.
Mammography Quality Standards Act MQSA requirements – Energy—save mode is built-in for use during periods of inactivity, use on-demand wake up for after—hours imaging, and preset wake-up time to meet your needs.
When set to ON, a one pixel frame is printed around each image.
Plus, the DryView is a true dry laser imager. Attention Message Area — Displays system status and error messages.
Drygiew Next, as required, to select Contrast. The Optics module will need to be re–initialized. DICOM print service class provider. Press the appropriate Format Select button to select the desired format number of images printed on a film.
The CR System hard disk should be capable of storing approximately 2, images on-line for rapid recall. P The supply door failed to open. Imager did not initialize in the correct sequence.