The EC help provides an easy access to the information related to the use of Keysight EC Network Analyzer. Pressing Help key on the front panel. EB/EB, please see the Programming Manual. • Installation and Quick Start Guide (Part Number: Ex1, attached to optional ABA). This manual . The EC complies with INSTALLATION CATEGORY II as well as The Service Manual is a guide to servicing the EC ENA Series Network Analyzer .
|Published (Last):||8 October 2017|
|PDF File Size:||5.74 Mb|
|ePub File Size:||5.64 Mb|
|Price:||Free* [*Free Regsitration Required]|
The complete process takes less than 5 minutes to finish.
The boot option is displayed. Click Cal Kit, then select the calibration kit.
Instructions contained in the warnings must be followed. A warning message is displayed. Component replacement and internal adjustments must be made by qualified maintenance personnel. A message dialog box appears after the completion of the Recovery Image Setup procedure. The procedure consists of the following items. DO NOT Service or Adjust the Instrument Alone – Do not attempt internal service or adjustment unless another person, capable of rendering first aid and resuscitation, is present.
Test and Measurement Call Center tel 1 fax 1 China: Push the button under the lock and remove the hard disk. No part of this document may be photocopied, reproduced, or translated to another language without the prior written consent of the Agilent Technologies. Types of system recoveries The following 2 types of system recoveries are available.
E5071d anything is missing, contact the nearest Agilent Technologies office. If you perform the following procedure incorrectly, a message asking you whether to return to the previous registration e501c and perform the registration appears. The Confirm Device Removal message box appears. Replacement Procedure Step 1. Turn the key to the left until it is locked. All rights are reserved. Select the C drive and D drive. Click Port x Manuwl to start the calibration measurement x denotes the test port to which the standard is connected.
Follow the procedure described below to calibrate the touch screen. This calibration makes it possible to perform measurements with the highest possible mwnual.
If you press F11 twice or more and a screen which is not mentioned in this manual appears, reboot the instrument and restart the process from the first. A message box appears after the completion of recovery process.
You can also contact one of the following centers and ask for a test and measurement sales representative.
Inset the key in the lock and turn anti-clockwise. The error correction function will also be automatically enabled. Turn ON the EC. CN or below Step 1. Do e50711c replace components with the power cable connected. Out-position of push-button switch.
Agilent Technologies EC Instruction manual |
CN or below This section describes how to return the contents of the C drive to the factory state. The User back up function is not available. The following screen appears during backup.
Such noncompliance would also violate safety standards of design, manufacture, and intended use of the instrument.
Test and Measurement Call Center tel 1 fax 1 Canada: To avoid injuries, always disconnect the power and discharge circuits before touching them.
Atec Agilent-E5071C User Manual
Factory recovery Returns the contents of the C drive to the factory state. If the hard disk failed and has been replaced after purchase, the state when the replacement was performed is recovered. The dialog box disappears automatically after a few seconds. Also, never save the contents of the drive to the Mmanual, D or F drive. Select the destination folder and input a file name, then click Next button.
Agilent Technologies assumes no liability for the customer’s failure manusl comply with these precautions. After a while, the EC restart automatically. When the recovery is finished, EC restarts. The touch screen calibration screen Figure appears.
Press any key to continue. Reconfiguring Touch Panel Settings Step 1. For more information, see Modifying Calibration Kit Definition. After restart, the screen for initial registration appears. NOTE With no operation on the touch screen calibration screen for a preset time, it automatically closes and the previous measurement screen reappears.
Figure Recover Factory Backup ecse Step 8. Files you created using the save function files in the D drive are not affected, but we recommend backing them up before executing system recovery for precautionary purposes.
A confirmation dialog box is displayed.